CONTENTS. Service Manual ECG C Contents The CE mark is applied only to the ECGL/M/N Electrocardiograph. This equipment complies. Service Manual ECG Composition. · To order a replacement assembly above, use the Code No. · To order a replacement component inside an. View Homework Help – Nihon_Koden_Cardiofax_ECG_-_Repair_manual from BA at Vietnam National University, Ho Chi Minh City. ECGL.
|Country:||Central African Republic|
|Published (Last):||18 March 2014|
|PDF File Size:||6.78 Mb|
|ePub File Size:||5.94 Mb|
|Price:||Free* [*Free Regsitration Required]|
Clean the thermal head. For details, refer to the instruction manual for the MRI. Remove the two M3 binding head screws washers which fasten the thermal head to the thermal head unit. After a system test is completed, you can execute other system test without exiting the System Test mode. To re-enter the system test number, do either of the following: When the cardiograph operates on battery power and large leakage current is input from the connected external instrument, ground the cardiograph or use an isolation transformer for the external instrument.
This will help us to promptly attend to your needs. Check that the patient cable is firmly connected to the electrodes and instrument. After quitting each test, the instrument returns to the standby mode eg entering the system test number. Not exactly what you want?
Cleaning and Lubricating Schedules To maintain the performance of the instrument, the parts listed in the table below must be regularly cleaned or lubricated. Disinfection and cleaning To disinfect the outside surface of the instrument, wipe it with a nonabrasive cloth moistened with alcohol. Before measurement, check that the 9620k plate is correctly attached to the patient and ect that the cardiograph operates correctly when using with the ESU. The battery is recyclable.
Before connecting or disconnecting any cables, turn off the instrument, unplug the AC power cord from the instrument and remove the battery pack. At the end of its useful life, under various state and local laws, it may be illegal to egc of this battery into the municipal waste stream.
To make the check jig, shortcircuit the pins as shown below.
Failure to follow this warning will cause serious electrical burn where the electrode is attached and damage the cardiograph due to discharge energy when defibrillation is performed.
Patient cable connector 1. Remove the power board from the lower casing. Turn the power off.
Nihon Kohden ECG – ECG
Signal line Ground line 4. When another type of power cord especially 2-prong power cord is used, this may cause electrical shock to the patient and operator. Attach all tips of each electrode lead to the input check jig. No electrode is attached to the patient or the RF RL electrode is not attached to the patient.
The newly entered numbers are automatically printed. If none of the actions solve the problem, contact your nearest Nihon Kohden 962l0.
The date and time is backed up with the lithium battery in a real time clock IC on the ECG control board. In addition, service, technical modification or any other product change performed by someone other than NKC or its authorized agents without prior consent of NKC may be 9620ll for voiding this warranty. Power off time out. When performing MRI tests, remove from the patient all electrodes which are connected to this cardiograph.
ECG Machines For Sale
Normal The LCD unit is faulty. Sound volume can be changed properly. Check with your local officials for proper disposal procedures. Remove the two M3 pan screws with 920l washers which fasten the motor sensor board to the motor assy and remove the motor sensor board.
Do not charge the deteriorated battery pack. Replace the ECG control board.
Preparation A locally made check jig is required for the test.